# A galvanometer with a resistance of 75Ω produces a full scale deflection with a current of 5 mA. How can this galvanometer be converted into an ammeter which has a range of 0-5A?

To convert a galvanometer into an ammeter, we need to add a shunt resistor in parallel with the galvanometer. Here’s how we can do it step by step:

**Step 1: Understand the given parameters**

- Resistance of the galvanometer, Rg=75Ω

- Full scale deflection current of the galvanometer, Ig=5mA=5×10−3A

- Desired maximum current for the ammeter, I=5A

**Step 2: Determine the current through the shunt resistor**

The total current I flowing through the circuit is divided between the galvanometer and the shunt resistor. The current through the galvanometer is Ig, so the current through the shunt Is can be expressed as:

Is=I−Ig

Substituting the values:

Is=5A−5×10−3A=4.995A

**Step 3: Apply the voltage equality in parallel**

Since the galvanometer and the shunt are in parallel, the voltage across both is the same. The voltage across the galvanometer Vg can be expressed as:

Vg=Ig×Rg

Substituting the values:

Vg=(5×10−3A)×(75Ω)=0.375V

**Step 4: Write the voltage equation for the shunt resistor**

The voltage across the shunt resistor Vs can also be expressed as:

Vs=Is×Rs

Where Rs is the resistance of the shunt. Since Vg=Vs:

Ig×Rg=Is×Rs

Substituting the known values:

0.375V=(4.995A)×Rs

**Step 5: Solve for the shunt resistance Rs**

Rearranging the equation to find Rs:

Rs=0.375V4.995A≈0.075Ω=75mΩ

**Conclusion**

To convert the galvanometer into an ammeter with a range of 0-5 A, a shunt resistor of approximately 75mΩ should be connected in parallel with the galvanometer.

---

**Step 1: Understand the given parameters**

- Resistance of the galvanometer, Rg=75Ω

- Full scale deflection current of the galvanometer, Ig=5mA=5×10−3A

- Desired maximum current for the ammeter, I=5A

**Step 2: Determine the current through the shunt resistor**

The total current I flowing through the circuit is divided between the galvanometer and the shunt resistor. The current through the galvanometer is Ig, so the current through the shunt Is can be expressed as:

Is=I−Ig

Substituting the values:

Is=5A−5×10−3A=4.995A

**Step 3: Apply the voltage equality in parallel**

Since the galvanometer and the shunt are in parallel, the voltage across both is the same. The voltage across the galvanometer Vg can be expressed as:

Vg=Ig×Rg

Substituting the values:

Vg=(5×10−3A)×(75Ω)=0.375V

**Step 4: Write the voltage equation for the shunt resistor**

The voltage across the shunt resistor Vs can also be expressed as:

Vs=Is×Rs

Where Rs is the resistance of the shunt. Since Vg=Vs:

Ig×Rg=Is×Rs

Substituting the known values:

0.375V=(4.995A)×Rs

**Step 5: Solve for the shunt resistance Rs**

Rearranging the equation to find Rs:

Rs=0.375V4.995A≈0.075Ω=75mΩ

**Conclusion**

To convert the galvanometer into an ammeter with a range of 0-5 A, a shunt resistor of approximately 75mΩ should be connected in parallel with the galvanometer.

### Similar Questions

A galvanometer coil has a resistance of 50Ω and the meter shows full scale deflection for a current of 5 mA . This galvanometer is converted into a voltmeter of range 0 - 20 V by connecting

View SolutionA moving coil galvanometer, having a resistance G, produces full scale deflection when a current Ig flows through it. This galvanometer can be converted into

(i) an ammeter of range 0 to I0(I0>Ig) by connecting a shunt resistance RA to it and

(ii) into a voltmeter of range 0 to V(V=GI0) by connecting a series resistance RV to it. Then,View SolutionA moving coil galvanometer has a coil of resistance 59Ω . It shows a full scale deflection for a current of 50 mA. How will you convert it to an ammeter having a range of 0 to 3 A?

View Solution(a) Briefly explain how a galvanometer is converted into an ammeter.

(b) A galvanometer coil has a resistance of 15Ω and it shows full scale deflection for a current of 4 mA. Convert it into an ammeter of range 0 to 6 A.View SolutionA galvanometer coil has a resistance of 15Ω and the meter shows full scale deflection for a current of 4mA. How will you convert the meter into an ammeter of range 0 to 6A?

View SolutionA galvanometer having a coil resistance of 60Ω shows full scale defection when a current of 1.0A passes thoguth it. It can vbe convered into an ammeter to read currents up to 5.0A by

View SolutionA galvanometer coil has a resistance of 12Ω and the meter shows full scale deflection for a current of 3mA. How will you convert the meter into a voltmeter of range 0 to 18V?

View SolutionA moving coil galvanometer of resistance 20Ω gives a full scale deflection when a current of 1mA is passed through it. It is to be converted into an ammeter reading 20A on full scale. But the shunt of 0.005Ω only is available. What resistance should be connected in series with the galvanometer coil?

View SolutionA moving coil galvanometer of resistance 20Ω gives a full scale deflection when a current of 1mA is passed through it. It is to be converted into an ammeter reading 20A on full scale. But the shunt of 0.005Ω only is available. What resistance should be connected in series with the galvanometer coil?

View SolutionA galvenometer with a coil of resistance 12.0Ω shows full-scale deflection for a current of 2.5mA. How will you convert the meter into

(a) an ammoter of range 0 to 7.5A?

(b) a voltmeter of range 0 to 1.0V?View Solution